3M DYNATEL 965DSPA ANALYZER WITH WB/TDR 6.0 REPAIRED Refurbished units. Software version 6.00
Rugged and portable for harsh environments, Dynatel 900 Series test sets offer the right mix of functions, from loop diagnostic routines to fault locating, transmission testing and combination test/terminal capabilities. The Dynatel 965DSP Subscriber Loop Analyzer is a microprocessor-controlled integrated test set that provides full-featured POTS and fundamental ISDN testing, including fault location and repair verification on twisted-pair, quad and coaxial cable. (includes capacitance bridge, resistance bridge and Time Domain Reflectometer (TDR) functions).The Dynatel 965DSP subscriber loop analyzer executes a wide range of individual tests, or performs automatic test routines as specified by the user to categorize and sectionalize problems. The hand-held 965DSP test set is housed in a lightweight, ergonomically designed case for portability and ease of handling. The unit is weather-resistant and weighs only 3.3 pounds
Specifications for 965DSP Subscriber Loop Analyzer
Auto Test to aid in identifyng-solving problem, TDR Single and Dual Trace 3-30,000 ft, Send and receive 1200 kHz Tone, Load coil count 0-4, Measures voltage 0-250 VAC, 0-300 VDC, Current 0-110 mA DC, Resistance 0-990 mega ohms, Opens 0-100 k feet, RFL 0-20 mega ohms, Loss (& Frequency @ 600 ohms ZIN) -40 to +10 dBm 200 Hz-19.9 kHz, Wideband loss(@ 135 ohms ZIN) -40 to +10 dBm, 20 k-200 kHz, ISDN 40 kHz, Tone output (@ 600 ohms ZIN) 200 Hz-19.95 kHz , Caller ID Date-time-number-name, Longitudinal balance 62 to 40 dB, Noise metallic 0 to 50 dBr nc, Noise to ground 40 to 100 dBrnc, Dial mode DTMF or Pulse, 1.2 amp-hour NiMH battery pack, 50% duty cycle-12 hours no backlight-4 hours with,
Size H x W x D in. 10.0 x 4.0 x 2.5 , Cord length ft. 5, Net weight lbs. 3.3, Operating temperature 0° to 140°F , Storage temperature -40° to 165°F, Relative humidity 0 to 95%, non-condensing, Operating altitude 0 to 16,600 ft., Display in. 2.5 x 2.5 , 192 by 192 pixel resolution, backlight
Rechargeable Battery units
This unit uses a sealed battery pack inside.
This battery pack must be charged on a regular basis to keep the life of the batteries optimal. Letting the batteries totally discharge will drastically shorten the life of them.
Plug the unit into 120VAC using the supplied charging pack - charge for a minimum of two hours or as needed.
Proceed with proper operation of the unit.
Stored results 100 results of each, minimum
(Autotest and TDR)